- Who is Sysmex ?
- Quality and Sustainability
- News & Events
- In-Vitro Diagnostics
- Oncology & Pathology
- Industry & Research
- Laboratory Equipment
- Environment & Safety
Percentage of hypo-haemoglobinised red cells (HYPO-He) and hyper-haemoglobinised red cells (HYPER-He)
What are HYPO-He and HYPER-He used for?
Together with other parameters, HYPO-He is used in RBC indices to distinguish between iron defiency anaemia and thalassaemia.
The use of HYPO-He as well as MicroR is recommended in nephrology guidelines such as the European Best Practice Guidelines (EBPG), National Kidney Foundation Kidney Disease Outcome Quality Initiative (NKF KDOQI).
How are HYPO-He and HYPER-He determined?
HYPO-He and HYPER-He are parameters analysed in the reticulocyte (RET) channel. They are derived from the haemoglobin content of all mature RBC (RBC-He), which can be calculated based on the high-angle forward scatter (FSC). The FSC signal is converted into picograms (pg) using a proprietary algorithm and in healthy individuals the resulting RBC-He value is comparable to the mean corpuscular haemoglobin (MCH) value. Subsequently, this RBC-He value is used to determine HYPO-He and HYPER-He.
HYPO-He is the percentage of RBC with cellular haemoglobin content lower than 17 pg, whereas HYPER-He is the percentage of RBC with cellular haemoglobin content higher than 49 pg.
HYPO-He and HYPER-He are diagnostic parameters on XN-Class analysers that have the RET channel.*
*In the XT-4000i and XE-5000 analysers these are research parameters that are derived from the RET channel and named %HYPO-He and %HYPER-He. Research parameters should not be used for in vitro diagnostics.